Serpens South Trial E4 Flags
Revision as of 14:00, 30 August 2012 by Manferna
2012-08-25 c0924.4E_88Serpen.4 5.5 B+ Comments: NODE_NOT_FOUND [2012-08-25 21:21]TP : Subtracted time when source was below 30 degrees elevation.
* C8 out * C23 no coherence * 1743-038 observing at low elevation (28, 23, 18 degrees) in last hour. Some 6-m gains go way up. Obs said they did this to fill a gap and won't charge that hour.
* The partially observed top box looks funny. It has the full N-S extent, but not the full E-W extent. How is it being mapped??? It was a problem in the OTF mosaic parameters. Fixed it.
* C17 had high Tsys when observing 3c279 but the data still can be used with bpass calibration purposes * C7 had a phase jump between t(05:42:42,06:43:43) * rms cont: 12/10
select='ant(8,23)' #no coherence select='ant(17),source(3c279)' #very high amp select='ant(9,14),time(06:48:53,08:00:00)' #high gains
uvbflag select="source(3c279,mars),ant(13,14,15)" # useful data for calibration purposes select='ant(8,23)' # out and uncorrelated select='ant(7),time(05:42:42,06:43:43)' # phase jump