L1451 Trial 1SLDE Flags
Revision as of 16:53, 7 March 2013 by Sstorm
2013-02-07 c0924V.2D_88B1E.SL.1 7.4 A Comments: [2013-02-08 16:08]WSM : C19 RX bad. Radio pointing offsets uncertain (e.g., some BIMA and SZA antennas affected). Need to apply baseline solution in post-processing. Weather was OK.[ Mon Feb 11 22:10:17 2013 UT ] S. Warren : see SL trial.[ Mon Feb 11 22:15:08 2013 UT ] S. Warren : see SL trial.
* Need baseline solutions to be applied in processing. DONE MARCH 7TH. * (These antennas still do not have good bl solns: C8, C9, C14, C15, C18, C19 --- old) * Radio pointing on 3.5-m very suspect -- flag C17, C18, C21, C22 due to amp drift on 3c84
cont rms: box1~7.5 mJy/bm ; box2~9.5 mJy/bm
apfile = 'antpos.130217' NEEDED! select='ant(19)' #bad Rx state select='ant(17,18,21,22)' #bad pointing select='source(3c111),time(01:08:50,01:09:30)' #bad amp select='source(3c84),time(22:35:00,22:54:30.0)' #bad amp