Serpens South Trial ESL13b Flags
2013-08-14 c0924I.4E_88Serpen.SL.13 4.2 A+ Comments: NODE_NOT_FOUND [2013-08-14 11:28]TH : c23 low coherence throughout the track. Passband calibrator was observed at a low elevation so several antennas were shadowed. c22 had tertiary issues. 74% time charged. 2013-08-14 c0924I.4E_88Serpen.WB.13 4.2 C Comments: NODE_NOT_FOUND [2013-08-14 11:28]TH : True weather grade was A+ but we failed the track to charge only for the SL time.
-c13,c14,c15: did not observe 3c279 -c8: did not observe mwc349 -We are using REFANT=c11 -c20: high tsys (450K vs 200K) -c22: has a jump in Tsys in the middle of the track (from 150K to 600K), so that most part of the track had high Tsys. -c23: uncorrelated -RMS noise: continuum, box1/2: 7.9/8.2 mJy/beam -RMS noise: HCO+ (w4), box1/2: 0.86/0.85 Jy/b