Serpens South Trial E9 Flags
2012-09-17 c0924.4E_88Serpen.9 5.0 A Comments: NODE_NOT_FOUND [2012-09-17 18:29]D.Fisher : data on C22 may not be trustworthy.
-MARS used as bpass calibrator. -Some antennas slightly shadowed but not a big issue in this occasion. -Weather improving from previus tracks. -C20, C22: offline. -C17 and C18 have consistently higher tsys, but this is not an issue for the SNR of the final maps. - RMS cont: 9/10 mJy/b - RMS HCO+: 0.69/0.74 mJy/b -Single Dish: window 4 on C2 seems bad.
-Used C8 for refant. -C10 has bad RxState for start of track. -cont rms: box1~8.5 mJy/bm; box2 too partial -HCO+ single channel rms: box1~680 mJy/bm
* select="ant(20,22)"# offline
select='ant(20,22)" # not in array select='ant(10),time(23:00:00,00:59:28)' #rxstate bad at start of track