Serpens South Trial E2 Flags
2012-08-23 c0924.4E_88Serpen.2 5.2 C Comments: NODE_NOT_FOUND [2012-08-23 11:14]TP : First two hours before rain storm hit may be useable.
* Tau ~ 2.8. RMSpath ~ 580. Worsen during the second half of the track. * Only box 1 was observed. * Observed 0102+584 as bandpass calibrator (it has a flux of ~4.9Jy at 3mm). Although it's useful for the continuum windo, it was not enough to do bpass on narrow windows. * Observed MWC349 as fluxcal calibrator. The flux for 1743-038 was of 3.37, about 15-20% below the flux measured during the Serpens Main tracks with other flux calibrators. * C7 was missing. * C17 was uncorrelated for the first half of the track ~(02:30,04:30). * For the narrow bands, I achieved to do the bpass calibration using the noise source, although the correction is not so good. * RMS continuum: 0.11 Jy/beam * RMS window 4 (empty channels): 1.1 Jy . km/s /beam
* uvcheck var=systemp range=0,1000 # outliers * select="ant(7,17)" # missing and uncorrelated antennas * select="time(04:30:00,07:12:00),-source(mwc349,0102+584)" # bad weather