Serpens South Trial E10 Flags
2012-09-16 c0924.4E_88Serpen.10 3.3 A Comments: NODE_NOT_FOUND [2012-09-16 23:35]D.Fisher :
* Some higher gains at the very start, but not being that picky. * cont rms: box2~8 mJy; box1~9.1 mJy * HCO+ single channel rms (NOISE source BP calibration used): box2~640 mJy; box2~768 mJy
* I found small drop on MARS amplitude (1min off) and the phase jump causing the higher gains at the beggining. * c9,c13,c21,c22: were shadowed when observing MARS --> uvbflag solved the problem. * RMS cont: box2/1: 8/9 mJy/b * RMS HCO+ 1 channel (MARS as bpass cal): box 2/1: 0.62/0.68 Jy/b; * SDish: win4, C5 --> bad
* select='ant(20)' #not in array
* select='ant(20)' #not in array * select="ant(4,5,6),time(00:02,00:03)"# amplitude drops for 1 min * select="time(00:06:00,00:07:10)"# phase jump causing higher gains * uvbflag vis=uvdata mask=7,8 flagval=unflag select="source($bpcal,$fxcal),ant(9,13,21,22)"