Serpens South Trial DSL3 Flags
2012-10-07 c0924I.4D_88Serpen.SL.3 5.9 A- Comments: NODE_NOT_FOUND [2012-10-07 22:51]CA & SS : Looks good. Calibrator below 30 degrees for last half hour, -0.5 hr. 2012-10-07 c0924I.4D_88Serpen.WB.3 6.4 C+ Comments: NODE_NOT_FOUND [2012-10-07 22:52]CA & SS : WB. Calibrator below 30 deg last half hour.
2012-10-07 c0924I.4D_88Serpen.SL.3 6.5 A- Comments: [2012-10-07 22:51]CA & SS : Looks good. Calibrator below 30 degrees for last half hour, -0.5 hr. [ Thu Nov 1 23:26:50 2012 UT ] NV : Low elev; C17,C23 shadow at end. C22 phase jump at 2UT. High Tsys on all ants at start.
- Data looks good. - When applying bootflux, note that when 1743-038 is below 30degrees, the estimated fluxes are lower than when it is at 40degrees. I have fixed it flux at 3.6 (which is the obtained when observing it at 40degrees). - I use 3c279 to bpass the continuum and MARS for the narrow bands (3c279 final spectra looks terrible on those narrow bands, dont know why). MARS is barely resolved but still can be used for bpass purposes. * RMS cont, box1/2: 8/8 mJy/beam (map not completed in top box, no WB added yet) * RMS HCO+, box1/2: 0.46/0.43 Jy/beam * SD: apparently ok. However, window 4 has really low SNR so keep an eye on it!! * WB, box1/2: 5/5 mJy/beam
* 3c279 has really high tsys and vector avg amp
* cont rms: box1~8 mJy/bm ; box2~7.3 mJy/bm * HCO+ single channel rms: box1~640 mJy/bm ; box2~567 mJy/bm
select="source(3c279)" #bogus/high tsys