Serpens South Trial DSL23Flags
2012-11-18 c0924I.4D_88Serpen.SL.23 0.0 C Comments: NODE_NOT_FOUND [2012-11-18 14:10]CRG : Bad weather. 2 antennas missing. Several antennas show periods of Tsys > 5000K. I-Bandpass phases all over the place. No time charged
* antennas in and antennas out. what were they doing??
* c8,c10: no bandpass calibrator * c6: no fluxcal calibrator * c16: had higher phase gains than the other 3.5m antennas * Astronomical bandpass could not be achieved for narrow bands. * Cont RMS box1/2: 9/12 mJy/b * HCO+ RMS box1/2: 1.1/1.5 Jy/b (high than in other tracks) * SD: apparently without problems.
tsys=500 #lots of whacky tsys select="ant(14),time(23:00:00,21:35:54.0)" # low amp
select="ant(8,10,16)" # lack of bandpass calibrator, c10 almost missing track select="ant(1,5),time(00:53,00:58)" # high gains select="ant(5),time(01:47,01:53)" # high gains select="ant(5),time(01:59:30,02:04:30)" # high gains select="ant(14),time(23:05,23:10)" # high gains