Serpens South Trial DSL10 Flags
2012-11-02 c0924I.4D_88Serpen.SL.10 6.1 A- Comments: NODE_NOT_FOUND [2012-11-02 20:28]jmc : C5 had drive problems at the beginning of track. C6 has high system temperatures due to failing refridgerator. The drive box on C5 was replaced with the drive box on C6. 2012-11-02 c0924I.4D_88Serpen.WB.10 6.1 C+ Comments: NODE_NOT_FOUND [2012-11-02 20:29]jmc : Giving the track a dummy grade since the SL track was graded.
* cont rms: box1~7.3 mJy/bm ; box2~8.1 mJy/bm * HCO+ single channel rms: box1~635 mJy/bm ; box2~638 mJy/bm * gains a bit high (~1.3) consistently on some 3.5-m
* C16,C18,C19: higer gains (~1.1 for me) * cont rms (1/2): ~9/9 mJy/b * HCO+ rms (1/2): 810/840 mJy/b (channels 147-150) * SD: remove C5 & C6 due to partial track and noise * WB: -baseline 1-4: ripple in amplitude -several bands show steep phases -final map seems really noisy --> probably not worth to use this -rms: 20 mJy/b
select="ant(5)" # it has not observed any calibrator