Serpens South Trial DESL2Flags
2013-02-11 c0924.4D_88Serpen.SL.2 2.6 A- Comments: NODE_NOT_FOUND [2013-02-11 19:51]A Schruba : Good weather. Bad antennas are: C5 intermittend bad locking, C2, C9, C15 C18, C19, C22 bad pointing; C14 out of array. 50% time charged.
* Decent weather track. C14 out of the array. * Applied new baseline solutions (antpos.130210). * C2, C18, C19 & C22 are uncorrelated. C19 shows steep slope in phase when plotting baseline spectra. * C4 shows bad amplitudes at the beggining of the track possibly due to bad pointing? * C8 & C19 show ripples in amplitude. * C15 did not observe 3C279 (even after uvbflag). * C16 shows higher gains at around ~1.2. * The overlapping region between the 2 boxes was no mapped. * cont RMS box1/2: 10/11 mJy/b * HCO+ cont bo1/2: 0.95/1.10 Jy/b * WB: different gains (and high) than for the SL track ??? --> data may not be useable
* I don't see why C2 needs to be tossed for the whole track Seems like the pointing was nailed down after ~1 hr into the track * Don't see problems with C5, C9, C15 that obs notes * BP with noise
* Flux of 1743-038 ~3.9 Jy * cont RMS box1/2: 10/13 mJy/b * HCO+ single channel rms box1/2: 0.8/1.0 Jy/b
select="ant(2,8,14,18,19,22)" select="ant(4),time(14:43,15:43)" select="time(06:10,06:15)" #ra/dec failure
select="ant(18,19,22)" # not correlated select="ant(14)" # not in array select="ant(8)" # amp ripples select="ant(2),time(14:00:00,15:43:12)" # bad ptng select="ant(4),time(14:00:00,15:43:12)" # bad ptng