Serpens South Trial DE1Flags
2013-02-16 c0924.4D_88Serpen.1 4.1 A Comments: NODE_NOT_FOUND [2013-02-16 18:11]Karto : Ant 14 and 17 show high tsys (ant 14 looks ok for the first part of the track). Data for ants 15 and 19 not present, 23 shows high gains.BP solutions noisy for 3 windows, likely due to narrow bandwidth. 0.5 hours deducted from total time due to high gains (between 16:00h-16:30h), 1 hour deducted due to 4 antennas being out of service (14, 15, 19, 50% of 9, 50% of 23). Weather grade A+ (though likely close to A/A-, phase monitor was out of service).
* C19: outline * C9, C14: second half of the track --> apparently Serpens not observed * C17: higher scatter in amplitude * C23: uncorrelated * Problem with some integrations (bad ra/dec values) * only observed the bottom box (box 2)
* cont RMS box1/2: -/5 mJy/b * HCO+ cont bo1/2: -/0.51 Jy/b
* 1743-038 flux ~3.9 Jy
select="ant(19,23)" select="ant(12),time(15:23,15:28)" # high gains select=time(16:43:53,16:57:02) # bad ra/dec values
ra/dec zeroed out, potentially due to C14 dropping out at this point? select=time(16:43:53,16:57:02)