Serpens Main Trial 8E Flags
2012-07-19 c0924I.3E_88Serpen.8 3.2 C Comments: NODE_NOT_FOUND[ Thu Jul 19 09:21:30 2012 UT ] Observer : Running checksafe. [2012-07-19 11:19]LR & VR : The script crashed earlier because of an antenna collision; no time has been subtracted to the project. Despite no fluxcal was observed during the track, these data might be usable if the flux on 1743-038 could be measured on another track of this same project.
-tau: 0.65-0.80 -rms path: 300-600 -No flux calibrator observed (3.88 used for 1743-038, as measured the previous day) -It seems only box 2 was observed (?) -C5 had TSYS problems and there were a couple of peaks in tsys at about 5.05, 5.25, and 7.05 to 7.35 -Last 2 integrations on serpens were not surrounded by a calibrator observation -C16,C18 and C19 had higher gains and I flagged C18 and C19 since they're the outermost antennas and do not contribute to the shorter baselines
* box 3, 2, and partial 1 were observed * there was a collision in the middle of box 1 * something is screwy with my narrow band calibration -- seeing lots of spikes in amp with uvspec * the wideband looked fine: got cont rms ~11/11/9
* select="ant(5,18,19)" # high tsys and gains * select="time(07:05:00,07:35:00)" # no calibrator observed after that * select="time(06:44:00,06:52:00)" # tsys peak * select="time(05:04:00,05:08:00)" # tsys peak * select="time(05:26:00,05:29:00)" # tsys peak
(still looking) select='ant(5)' #tsys select='time(05:58:59.5,06:03:01.0)' #bad wideband phase solution select='ant(18,21)' #bad in narrow band