Serpens Main Trial 6E Flags
2012-07-13 c0924I.3E_88Serpen.6 4.8 B Comments: NODE_NOT_FOUND [2012-07-13 10:34]AS & LR : Data look good overall. Track was run for 5.8 hours in total, but we did not subtract the last hour to the project because of very high opacity.
* John C took an independent look at this dataset on request of Lee * He recommends it gets redone * Lee will pass that info to NV once he is back from vacation * Based on manuel's flags, I get 13 mJy rms for continuum in all boxes * so if that is 2x worse than usual, getting charged for 1/4 time is fair (?)
*Bad weather track (tau~2 with peaks up to 10, rmspath:200-500) *2 peaks on tsys at ~6.20 and ~9.00 LST *C16 and C19 again have higher gains than the other antennas. *C9,C21 have not good data for Neptune (shadowed??) *Several antennas shadowed mainly while observing 1743-038 at the end of the track *The Tsys of this track is ~200 K above that of the others *I detected 2 of the main 3 continuum sources up to a 3-sigma threshold (~36 mJy/beam)
select="ant(16,19)" # higher gains and noise bandpass baselines select="time(06:15:00,06:30:00)" # tsys select="time(08:30:00,09:20:00)" # tsys select="ant(18),time(04:48:00,04:54:30)" # high gains select="ant(18),time(07:16:00,07:20:00)" # high gains