Serpens Main Trial 1E Flags
2012-07-04 c0924I.3E_88Serpen.1 4.2 B- Comments: NODE_NOT_FOUND [2012-07-04 10:30]DR : Bandpass solution is bad, probably because the bandpass calibrator is too weak. Grade was decreased to B-
* 1743-038 observed for BP possibly because it is brighter than the one "preferred" calibrator that were up at the start of the track. * We probably will need to use the noise source to bp calibrate this track, and the D array tracks that did not get 3c279.
* c10, c15: no neptune data * c13,c14: some missing integrations on 1743-038 at the end of the track * c21: 1743-038 possibly shadowed in the middle of the track * ovro-C21 baselines: low SNR for bpass
select='ant(4),time(06:15:51.8,07:02:01)' flagval=f select='ant(15),time(05:07:00,05:47:01.0)' flagval=f
select='ant(4),time(06:36:00,06:39:00),source(1743-038)' flagval=f # Tsys outliers on 1743-038 select='ant(10),time(06:15:51.8,07:01:12.0)' flagval=f # Jump in Tsys spans 1743-038 and Serpens integrations select='ant(15),time(05:00:00.0,05:46:13.5)' flagval=f # Bad tsys 2 first 1743-038 integrations select='ant(14),time(08:11:41.8,09:12:21.5)' flagval=f # shadowed calibrator??? select='ant(21),time(06:41:01.8,08:30:47.5)' flagval=f # Shadowed calibrator???