L1451 Trial 3SLDE Flags
2013-02-11 c0924V.2D_88B1E.SL.3 5.2 A- Comments: [2013-02-11 14:05]S. Warren : C18, C19, C22 bad pointings - not usable. C14 out of array. C8 claims to have been shadowed. Most likely this data is good and should be unflagged using UVBFLAG in Miriad. Please apply new baseline solution in offline data reduction. 82% time charged.[ Mon Feb 11 22:11:28 2013 UT ] S. Warren : see SL trial.[ Mon Feb 11 22:12:38 2013 UT ] S. Warren : see SL trial.
* Need baseline solutions to be applied in processin -- FIXED MARCH 4TH * These antennas did not have good bl solns: C8, C9, C14, C15, C18, C19 * Radio pointing on 3.5-m very suspect -- flag C18, C19, C21, C22 due to amp drift on 3c84 * C17 seems to have ptng improved cont rms: box1~7.5 mJy/bm ; box2~6.7 mJy/bm HCO+ single channel rms: box1~569 mJy/bm ; box2~503 mJy/bm
(before baseline corrections cont rms: box1~7.3 mJy/bm ; box2~9.0 mJy/bm
apfile='antpos.130217' select='ant(18,19,22)' #bad ptng select='ant(14)' #not in array select='ant(21)' #bad pointing select='ant(5),time(05:58:17.7,07:00:00),win(1)' #bad nb gains