B1s Trial 6E Flags
2012-07-21 c0924.1E_88B1S.6 10.9 B Comments: NODE_NOT_FOUND [2012-07-21 14:56]MH & VR : The last two hours of the track are unstable and probably not usable.
* below ~35 degrees elevation, had 6/23 antennas shadowed according to RTD. Is this right? * Peter says that the flagging/shadowing might not be working properly * Weather got bad last 3 hours. We should probably not get charged for those (? pre-inspection)
* I agree that the data since 18:00 is not good. Should not get charged. * Something is wrong with C5.
select='ant(5)' #tsys select='ant(21),time(09:00:00.0,10:47:37.5)' # low amp select='ant(21),time(12:18:03,12:45:52.0)' # low amp select='time(18:19:33.8,19:14:32)' # bad weather select='ant(16),time(19:55:38,21:00:00)' # high gain select='ant(18),time(19:55:38,21:00:00)' # high gain select='time(20:16:22.8,20:30:00)' # no phase cal end
NEW! select='ant(1,2,3,4),time(17:00:00,21:00:00)' # low amp NEW! select='source(3c84,B1S),time(10:00:00,11:00:00)' # bad phases
select=ant(5) # Bad tsys select=ant(1,2,3,4,7,8,9,10,11,12,13,14,15),time(18:30,19:10) # high amp gain select=ant(16,17,18,19,20,21,22,23),time(18:30,20:20) # high amp gain select=ant(18),time(15:30,15:40) # high amp gain select=ant(21),time(10:00,10:20) # low amp select=ant(21),time(12:15,12:35) # low amp